- Carlo Pancaro, Jasmine Purtell, Dana LaBuda, Leif Saager, Thomas T Klumpner, Timur Dubovoy, Baskar Rajala, Shubhangi Singh, Ruth Cassidy, Christie Vahabzadeh, Sean Maxwell, Virgil Manica, David M Eckmann, Jill M Mhyre, and Milo C Engoren.
- From the Department of Anesthesiology, University of Michigan, Ann Arbor, Michigan.
- Anesth. Analg. 2021 Apr 9.
BackgroundWhile flexible epidural catheters reduce the risk of paresthesia and intravascular cannulation, they may be more challenging to advance beyond the tip of a Tuohy needle. This may increase placement time, number of attempts, and possibly complications when establishing labor analgesia. This study investigated the ability to advance flexible epidural catheters through different epidural needles from 2 commonly used, commercially available, epidural kits.MethodsWe hypothesized that the multiorifice wire-reinforced polyamide nylon blend epidural catheters will have a higher rate of successful first attempt insertion than the single-end hole wire-reinforced polyurethane catheters for the establishment of labor analgesia. The primary outcome was a difference in proportions of failure to advance the epidural catheter between the 2 epidural kits and was tested by a χ2 test. Two-hundred forty epidural kits were collected (n = 120/group) for 240 laboring patients requesting epidural analgesia in this open-label clinical trial from November 2018 to September 2019. Two-week time intervals were randomized for the exclusive use of 1 of the 2 kits in this study, where all patients received labor analgesia through either the flexible epidural catheter "A" or the flexible epidural catheter "B." Engineering properties of the equipment used were then determined.ResultsFlexible epidural catheter "A," the single-end hole wire-reinforced polyurethane catheter, did not advance at the first attempt in 15% (n = 18 of 120) of the parturients compared to 0.8% (n = 1 of 120) of the catheter "B," the multiorifice wire-reinforced polyamide nylon blend epidural catheter (P < .0001). Twenty-five additional epidural needle manipulations were recorded in the laboring patients who received catheter "A," while 1 epidural needle manipulation was recorded in the parturients who received catheter "B" (P < .0001). Bending stiffness of the epidural catheters used from kit "B" was twice the bending stiffness of the catheters used from kit "A" (bending stiffness catheters "A" 0.64 ± 0.04 N·mm2 versus bending stiffness catheters "B" 1.28 ± 0.20 N·mm2, P = .0038), and the angle formed by the needle and the epidural catheter from kit "A" was less acute than the angle formed from kit "B" (kit "A" 14.17 ± 1.72° versus kit "B" 21.83 ± 1.33°, P = .0036), with a mean difference of 7.66° between the 2 kits' angles.ConclusionsThe incidence of an inability to advance single-end hole wire-reinforced polyurethane catheter was higher compared to the use of multiorifice wire-reinforced polyamide nylon blend epidural catheter. Variation of morphological features of epidural needles and catheters may play a critical role in determining the successful establishment of labor epidural analgesia.Copyright © 2021 International Anesthesia Research Society.
Knowledge, pearl, summary or comment to share?
You can also include formatting, links, images and footnotes in your notes
- Simple formatting can be added to notes, such as
- Superscript can be denoted by
- Numbered or bulleted lists can be created using either numbered lines
1. 2. 3., hyphens
- Links can be included with:
[my link to pubmed](http://pubmed.com)
- Images can be included with:
![alt text](https://bestmedicaljournal.com/study_graph.jpg "Image Title Text")
- For footnotes use
[^1](This is a footnote.)inline.
- Or use an inline reference
[^1]to refer to a longer footnote elseweher in the document
[^1]: This is a long footnote..