- Amrut K Rao, Souvik Chaudhuri, Tim T Joseph, Deependra Kamble, Gopal Gotur, and Sandeep Venkatesh.
- Department of Anaesthesia, Kasturba Medical College, Manipal University, Manipal, Karnataka, India.
- Anesth Essays Res. 2014 May 1; 8 (2): 243-6.
AbstractA well-secured endotracheal tube (ETT) is essential for safe anesthesia. The ETT has to be fixed with the adhesive plasters or with tie along with adhesive plasters appropriately. It is specially required in patients having beard, in intensive care unit (ICU) patients or in oral surgeries. If re-adjustment of the ETT is necessary, we should be cautious while removal of the plasters and tie, as there may be damage to the cuff inflation system. This can be a rare cause of ETT cuff leak, thus making maintenance of adequate ventilation difficult and requiring re-intubation. In a difficult airway scenario, it can be extremely challenging to re-intubate again. We report an incidence where the ETT cuff tubing was severed while attempting to re-adjust and re-fix the ETT and the patient required re-intubation. Retrospectively, we thought of and describe a safe, reliable and novel technique to prevent cuff deflation of the severed inflation tube. The technique can also be used to monitor cuff pressure in such scenarios.
Knowledge, pearl, summary or comment to share?
You can also include formatting, links, images and footnotes in your notes
- Simple formatting can be added to notes, such as
- Superscript can be denoted by
- Numbered or bulleted lists can be created using either numbered lines
1. 2. 3., hyphens
- Links can be included with:
[my link to pubmed](http://pubmed.com)
- Images can be included with:
![alt text](https://bestmedicaljournal.com/study_graph.jpg "Image Title Text")
- For footnotes use
[^1](This is a footnote.)inline.
- Or use an inline reference
[^1]to refer to a longer footnote elseweher in the document
[^1]: This is a long footnote..